Last update - 10/11/2006
The Multispectral Radiometer (MSR) System is shipped with the MSR program and initial configuration/calibration pre-installed in the Data Logger Controller (DLC). A copy of the installed MSR progam and initial configuration/calibration is also saved on the DLC memory card (initially located with the MSR software installation diskette inside the User's Manual). Additionally, a copy of the initial configuration/calibration settings and values are stored in a file on the diskette and in the MSR folder on your computer after the MSR software is installed from the diskette. The name of the at file is MSRnnn.cfg, where nnn equals the three digit serial number found on the tag on the MSR sensor head.
If you lost the MSR program and configuration/calibration from the DLC memory card or if you recently reconfigured or recalibrated your MSR System, you may save the MSR program and the newer configuration/calibration to the DLC memory card for quick reloading into the DLC at some later time.
The MSR configuration/calibration may also be saved to a pc file, from which it can later be restored, should that be necessary and the DLC memory card is not available or had it not been saved on the memory card.
Below, are instructions to store the MSR program and current configuration/calibration to the DLC memory card or to store the current MSR System configuration/calibration to a pc file.
|The memory card inserts to about 1/3rd of its length and you should feel a slight 'click' into place as it seats into the connector:|
|If the correct com port number is not shown in brackets, type it now|
If the above did not appear but the DLC Main Menu appeared intead and you wish to be storing the MSR Program and a new configuration/calibration you will need to Reload the MSR.BAS Program and Configuration/Calibration from the PC Files and possibly recalibrating (see the 'Calibration Instructions' section of the the Technical Support page) first before returning to the above steps.
|The memory card content index will be displayed|
If it does appear, move the write protect switch to the OFF position. See memory card differences for some pictures of where the write protect switch or slot is located.
If it does not appear then the switch is already in the off position. Contnue with the next step.
This completes the storing of program and configuration/calibration to the DLC memory card. You may return to the MSR Main Menu and put the DLC to sleep.
After the save has been completed, the CROPSCAN MULTISPECTRAL RADIOMETER SYSTEM MENU will appear.
This completes the saving of an MSR Configuration/Calibration to a PC file.
©2017 CROPSCAN, Inc.
all rights reserved